GC Flexceed Putty And Kit
Rubber Base Impression Material
Rs.1,350.00 – Rs.3,404.00
GC Flexceed Putty
|not available|| |
GC Flexceed Putty + Light Body Kit
|not available|| |
GC Flexceed Light Body refill
|not available|| |
GC Flexceed is a vinylpolysiloxane (addition silicone) impression material used for making accurate dental impressions with excellent reproducibility. It is available in Putty and light body consistencies to aid dentists make perfect impressions for fabrication of crowns, bridges, inlays, onlays and veneers. GC Flexceed accurately replicates the highest level of detail required for crown and bridge restorations. It helps make predictable impressions under all conditions. GC Flexceed is more hydrophilic and has excellent tear strength. In addition to capturing critical intricate details it also displaces moisture and resists deformation. The hydrophilicity of the material is also attributed to the low contact angle. It exhibits pseudo-plastic properties to enhance precision. Excellent wetting characteristics of the material ensure a hard gypsum cast with a smooth surface. GC Flexceed impression material exhibits good dimensional stability, that allows pouring of multiple casts up to two weeks. It also has superior tear strength properties in comparison with other VPS materials. The resistance to tear in the subgingival and interproximal areas is significant as they are thin and are susceptible to tear. As measured by the ‘Shark Fin Test’ GC Flexceed exhibits the highest ‘fin’ when compared to other VPS impression materials. This vindicates its excellent reproduction of detail and flow characteristics.
|Features||Excellent detail reproduction with two viscosities. Pseudo plastic properties for precision Superior tear strength properties Long term dimensional stability - 2 weeks Low contact angle Better hydrophilicity Compatibility with gypsum products Better wetting characteristics due to the low contact angle Can be cold sterilized|
PHYSICAL PROPERTIES: LIGHT BODY PUTTY Mixing Time 20 sec Total Working Time 2 min 1 min Minimum time in Mouth 3 min 3 min Elastic recovery More than equals 98% More than equals 98% Strain in Compression 2.5-3.5 % 1-2% Linear dimensional change Less than or equals to 0.20% Less than or equals to 0.20% Detail Reproduction 20 μm 75 μm Compatibility with Gypsum 20 μm 75 μm
Precautions to be taken: Care should be taken while extruding or mixing GC Flexceed, to avoid contact with the following materials. (catalyst for C Silicone Impression materials, Polysulfide impression materials, eugenol materials, sulphur, latex, oil, acrylates, glycerol and local anaesthesia- spray tube or ointment type.) Use plastic or polyethylene gloves while kneading the putty material. Do not use latex gloves. Do not interchange the caps or measuring spoons of the containers. Storage: Store at normal room temperature (15 – 25oC) and avoid direct sunlight. Shelf life – 3 years from date of manufacture.
|Directions To USE|
I. Single step putty wash impression STEP 1: Take equal quantities of FLEXCEED base & catalyst using the prescribed scoop STEP 2: Knead the FLEXCEED base & catalyst until a uniform blue color is achieved STEP 3: Load the FLEXCEED putty mix on the tray STEP 4: Inject the FLEXCEED light body onto the putty mix making sure that the nozzle remains immersed to prevent air entrapment STEP 5: Using the intra-oral tips, inject the FLEXCEED light body directly around the prepared tooth STEP 7: Final impression is easy to read due to the contrasting colour of FLEXCEED II. Impression for complete denture STEP 1: Apply GC tray adhesive STEP 2: Bourder moulding using FLEXCEED putty STEP 3: FLEXCEED light body is loaded onto the individual tray STEP 4: Completed upper impression Double step putty wash impression STEP 1: Prepared tooth for crown STEP 2: After applying GC tray adhesive, FLEXCEED putty Putty impression is taken is loaded on the stock tray STEP 3: Putty impression is taken STEP 4: Using the putty cutter, create a space for the light body STEP 5: Inject the FLEXCEED light body around the prepared tooth, then immediately re-seat the putty tray impression STEP 6: Final impression using FLEXCEED Dual Arch impression STEP 1: FLEXCEED putty is loaded onto a dual arch tray STEP 2: FLEXCEED Light Body is injected over the putty Final impression. then impression is taken
FLEXCEED KIT: PUTTY - 1x Base - 500g (278mL) 1x Catalyst - 500g (278mL) LIGHT BODY - 2x Cartridges - 78g (50mL) 8 Mixing Tips (Pink) 8 Intraoral Tips FLEXCEED PUTTY: 1x Base - 500g (278mL) 1x Catalyst - 500g (278mL) FLEXCEED LIGHT BODY: 2x Cartridges - 78g (50mL) 8 Mixing Tips (Pink) 8 Intraoral Tips